Showing all 11 results
Thermocouple Temperature Transmitters
Important Thermocouple Information
The most important thing to remember about the TC is it that it’s the temperature difference between ends that produces the charge imbalance that drives the small voltages, not the junction of the two dissimilar metals that form the thermocouple circuit.
While you can form your own TC circuit with different combinations of two conductors, standard TC types using specific metals and alloys are available that produce larger, stable, and predictable output voltages relative to applied thermal gradients. These standard types have their voltage over temperature tabulated in TC type tables relative to a cold junction at 0°C and 0.000mV.
These TC voltages could have been referenced to another temperature than 0°C, but 0°C was chosen because it is easily reproducible within ±0.2°C using a mixture or slurry of ice and water. By holding the cold junction at 0°C, the temperature of the other end directly corresponds to its thermoelectric voltage found by look-up in the standard TC type table.